BioScope Resolve. The Most Advanced AFM for Life Science Research. Andrea Slade, PhD. Senior Life Sciences Applications Scientist. JSPM Instructional Manual · SPM Training Manual – Bioscope II · SPM Training Manual – Multimode · User Manual for Electrostatic Force Microscopy. Magnetic Force Microscopy, and this manual is designed to serve as a supplement to All SPM Lab users must log on to the Bioscope II first in order to turn the.
|Published (Last):||3 September 2005|
|PDF File Size:||15.96 Mb|
|ePub File Size:||10.19 Mb|
|Price:||Free* [*Free Regsitration Required]|
BioScope™ Instruction Manual
For more information regarding this attachment, contact Digital Instruments. Basic Imaging Techniques 7.
Kowalski Read Me First! A la manuql de problemes occasionnels, certains utilisateurs ont experimentes des fuites du joint de la cellule liquide. Attach X-Y manipulator Attach the X-Y manipulator to the stage plate using three knurled thumb screws located on the right-underside of the stage plate. This closed-loop feedback control is the heart of the imaging portion of the control station.
The center frequency can be decreased to the point where the vibrational bioscopw reaches 90 percent of the maximum value. Always lower objective lenses by turning the focus knob before rotating the lens turret. Under the Motor menu select the Step Motor command.
To better understand what is being viewed when selecting different Data types, consider the diagram below: The NanoScope imaging system utilizes over parameters. The manual also contains a tab-indexed table of contents to bipscope searching times. Oscillations are an indication of too much feedback correction voltage sent to Z. There remains always some residual error which causes the system to approach, but not quite reach, the target state. If all goes well after re-engaging, a well formed image will begin to appear on the manuaal monitor.
The end of the scanner allows easy removal of the imaging probe assembly. The scanner houses the piezoelectric transducer.
Bioscope Manual for SWI
The Dimension head and the attached assembly may be detached from the BioScope and used as a stand-alone unit. When the assembly is removed, the optical port that normally transfers the image to the eyepieces will be accessible.
COM 2 is reserved. The desired operating amplitude depends on the sample, and other scanning conditions.
BioScope™ Instruction Manual |
The upperleft screw moves the laser spot up and down as gioscope on the computer screen detector graphic, and side to side long axis on the dark red laser spot screen directly on the SPM head. The Zoom In subcommand stretches the plot by decreasing the sweep width and shifting the center frequency value.
Plug the vacuum pump in and switch on. The positioners adjust the position of the mirror. It may be convenient to break several substrates from the manuao at one time. Note that this will result in an image that shows only large transitions in Z; this is normal.
Be aware of clearances between the AFM probe and the top of the sample whenever using the X-Y manipulator see Figure 3. The information in this manual biosdope furnished for informational use only, is subject to change without notice, and should not be construed as a commitment by Digital Instruments, Inc.
This allowed the tip to be biosco;e and lowered, keeping tip height constant over surface features and accommodating rougher samples. Bitte immer das Objektiv mit der Fokussiereinrichtung herunterfahren bevor der Objektivrevolver gedreht wird. On the Channel 2 panel, set Data type to Off to disable the secondary channel.
Warning statements identify conditions or practices that could result in bodily injury or loss of life. This section discusses adjustments required to improve force curves obtained after nioscope the microscope. Der Laser sollte niemals an die Mikroskopelektronik angeschlossen werden, wenn er nicht in der Halterung der ZVerschiebeeinheit installiert ist.
Continue to Zoom In and center the peak until the peak coincides with the vertical center line within about 10 Hz. Piezo oscillations typically cause high frequency wavy lines in the Real-time image. As the tip encounters a decline in the sample topology, the tip drops. But how were images produced? Regrettably, along with the increased sharpness buoscope the tip comes a slight increase in double tip effect experienced with the oxide sharpened process.